Subscribe to RSS
© Georg Thieme Verlag KG Stuttgart · New York
A magnetic internal mechanism for precise orientation of the camera in wireless endoluminal applications
submitted 5 November 2009
accepted after revision 3 March 2010
26 May 2010 (online)
Background and study aims: The use of magnetic fields to control operative devices has been recently described in endoluminal and transluminal surgical applications. The exponential decrease of magnetic field strength with distance has major implications for precision of the remote control. We aimed to assess the feasibility and functionality of a novel wireless miniaturized mechanism, based on magnetic forces, for precise orientation of the camera.
Materials and methods: A remotely controllable endoscopic capsule was developed as proof of concept. Two intracapsular moveable permanent magnets allow fine positioning, and an externally applied magnetic field permits gross movement and stabilization. Performance was assessed in ex vivo and in vivo bench tests, using porcine upper and lower gastrointestinal tracts.
Results: Fine control of capsule navigation and rotation was achieved in all tests with an external magnet held steadily about 15 cm from the capsule. The camera could be rotated in steps of 1.8°. This was confirmed by ex vivo tests; the mechanism could adjust the capsule view at 40 different locations in a gastrointestinal tract phantom model. Full 360° viewing was possible in the gastric cavity, while the maximal steering in the colon was 45° in total. In vivo, a similar performance was verified, where the mechanism was successfully operated every 5 cm for 40 cm in the colon, visually sweeping from side to side of the lumen; 360° views were obtained in the gastric fundus and body, while antrally the luminal walls prevented full rotation.
Conclusions: We report the feasibility and effectiveness of the combined use of external static magnetic fields and internal actuation to move small permanent intracapsular magnets to achieve wirelessly controllable and precise camera steering. The concept is applicable to capsule endoscopy as to other instrumentation for laparoscopic, endoluminal, or transluminal procedures.
- 1 Park S, Bergs R A, Eberhart R. et al . Trocar-less instrumentation for laparoscopy magnetic positioning of intra-abdominal camera and retractor. Ann Surg. 2007; 245 379-384
- 2 Scott D J, Tang S J, Fernandez R. et al . Completely transvaginal NOTES cholecystectomy using magnetically anchored instruments. Surg Endosc. 2007; 21 2308-2316
- 3 Ryou M, Thompson C C. Magnetic retraction in natural orifice transluminal endoscopic surgery (NOTES): addressing the problem of traction and countertraction. Endoscopy. 2009; 41 143-148
- 4 Wang Z, Wang L, Tang B. et al . Retraction by surface ferromagnetisation of target tissues: preliminary studies on feasibility of magnetic retraction for endoscopic surgery. Surg Endosc. 2008; 22 1838-1844
- 5 Lehman A C, Berg K A, Dumpert J. et al . Surgery with cooperative robots. Comput Aided Surg. 2008; 13 95-105
- 6 Hogle N J, Hu T, Allen P K, Fowler D L. Comparison of monoscopic insertable, remotely controlled imaging device with a standard laparoscope in a porcine model. Surg Innov. 2008; 15 271-276
- 7 Cadeddu J A, Fernandez R, Desai M. et al . Novel magnetically guided intra-abdominal camera to facilitate laparoendoscopic single-site surgery: initial human experience. Surg Endosc. 2009; 23 1894-1899
- 8 Carpi F, Galbiati G, Carpi A. Controlled navigation of endoscopic capsules: Concept and preliminary experimental investigations. IEEE Trans Biomed Eng. 2007; 54 2028-2036
- 9 Wang X, Meng M QH. A magnetic stereo actuation mechanism for active capsule endoscope. Conf Proc IEEE Eng Med Biol Soc. 2007; 2007 2811-2814
- 10 Valdastri P, Quaglia C, Susilo E. et al . Wireless therapeutic endoscopic capsule. Endoscopy. 2008; 40 979-982
- 11 Volke F, Keller J, Schneider A. et al . In-vivo remote manipulation of modified capsule endoscopes using an external magnetic field. Gastrointest Endosc. 2008; 5 AB121-AB122
- 12 Gheorghe C, Iacob R, Bancila I. Olympus capsule endoscopy for small bowel examination. J Gastrointest Liver Dis. 2007; 16 309-313
- 13 Simi M, Ciuti G, Tognarelli S. et al . Magnetic link design for a robotic laparoscopic camera. Journal of Applied Physics. 2010; 107 09B302
- 14 Ciuti G, Donlin R, Valdastri P. et al . Robotic versus manual control in magnetic steering of an endoscopic capsule. Endoscopy. 2010; 42 148-152
- 15 Tortora G, Valdastri P, Susilo E. et al . Propeller-based wireless device for active capsular endoscopy in the gastric district. Min Invas Ther Allied Technol. 2009; 18 280-290
- 16 Toennies J L, Webster R J. A wireless insufflation system for capsular endoscopes. ASME Journal of Medical Devices. 2009; 3 27514
- 17 Carta R, Lenaerts B, Thoné J. et al . Wireless power supply as enabling technology towards active locomotion in capsular endoscopy. Biosens Bioelectron. 2009; 25 845-851
- 18 Ciuti G, Valdastri P, Menciassi A, Dario P. Robotic magnetic steering and locomotion of capsule endoscope for diagnostic and surgical endoluminal procedures. Robotica.. 2010; 28 199-207
- 19 Carpi F, Pappone C. Magnetic manoeuvring of endoscopic capsules by means of a robotic navigation system. IEEE Trans Biomed Eng. 2009; 56 1482-1490
Scuola Superiore Sant’Anna
viale R. Piaggio 34