Subscribe to RSS
© Georg Thieme Verlag KG Stuttgart · New York
Robotic versus manual control in magnetic steering of an endoscopic capsule
submitted 29 May 2009
accepted after revision 24 November 2009
16 December 2009 (online)
Background and study aims: Capsular endoscopy holds promise for the improved inspection of the gastrointestinal tract. However, this technique is limited by a lack of controlled capsule locomotion. Magnetic steering has been proposed by the main worldwide suppliers of commercial capsular endoscopes and by several research groups. The present study evaluates and discusses how robotics may improve diagnostic outcomes compared with manual magnetic steering of an endoscopic capsule.
Materials and methods: An endoscopic capsule prototype incorporating permanent magnets was deployed in an ex vivo colon segment. An operator controlled the external driving magnet manually or with robotic assistance. The capsule was maneuvered through the colon, visualizing and contacting targets installed on the colon wall. Procedure completion time and number of targets reached were collected for each trial to quantitatively compare manual versus robotic magnetic steering (t-test analysis with P = 0.01). Then, through a set of in vivo animal trials, the efficacy of both approaches was qualitatively assessed.
Results: In ex vivo conditions, robotic-assisted control was superior to manual control in terms of targets reached (87 % ± 13 % vs 37 % ± 14 %). Manual steering demonstrated faster trial completion time (201 ± 24 seconds vs 423 ± 48 seconds). Under in vivo conditions, the robotic approach confirmed higher precision of movement and better reliability compared with manual control.
Conclusions: Robotic control for magnetic steering of a capsular endoscope was demonstrated to be more precise and reliable than manual operation. Validation of the proposed robotic system paves the way for automation of capsular endoscopy and advanced endoscopic techniques.
- 1 Iddan G, Swain P. History and development of capsule endoscopy. Gastrointest Endosc. 2004; 14 1-9
- 2 Moglia A, Menciassi A, Dario P, Cuschieri A. Capsule endoscopy: progress update and challenges ahead. Nat Rev Gastroenterol. 2009; 6 353-362
- 3 Van Gossum A, Munoz Navas M, Fernandez-Urien I. et al . Capsule endoscopy versus colonoscopy for the detection of polyps and cancer. N Engl J Med. 2009; 361 264-270
- 4 Wang X, Meng M. A magnetic stereo actuation mechanism for active capsule endoscope. IEEE Eng Med Biol Mag. 2007; August 2811-2814
- 5 Quirini M, Menciassi A, Scapellato S. et al . Feasibility proof of a legged locomotion capsule for the GI tract. Gastrointest Endosc. 2008; 67 1153-1158
- 6 Olympus, Inc., Tokyo, JAPAN. http://www.olympus.it/ Publication date: 8. December 2009
- 7 Volke F, Keller J, Schneider A. et al . In-vivo remote manipulation of modified capsule endoscopes using an external magnetic field. Gastrointest Endosc. 2008; 67 AB121-AB122
- 8 Valdastri P, Quaglia C, Susilo E. et al . Wireless therapeutic endoscopic capsule: in vivo experiment. Endoscopy. 2008; 40 979-982
- 9 Intuitive Surgical, Inc., Sunnyvale, USA. http://www.intuitivesurgical.com/patientresources/hospitals_doctors/index.aspx Publication date: 8. December 2009
- 10 Stereotaxis, Inc., St Louis, USA. http://www.stereotaxis.com Publication date: 8. December 2009
- 11 Carpi F, Pappone C. Magnetic maneuvering of endoscopic capsules by means of a robotic navigation system. IEEE Trans Biomed Eng. 2009; 56 1482-1490
- 12 Ciuti G, Valdastri P, Menciassi A, Dario P. Robotic magnetic steering and locomotion of capsule endoscope for diagnostic and surgical endoluminal procedures. Robotica. 2009; 156 49-58
- 13 VECTOR European Project. http://www.vector-project.com/ Publication date: 8. December 2009
- 14 Toennies J L, Webster III R J. A wireless insufflation system for capsular endoscopes. ASME Design of Medical Devices Conference, 2009. Poster Presentation. http://research.vuse.vanderbilt.edu/MEDLab/publications.htm Publication date: 8. December 2009
- 15 Park S, Bergs R A, Eberhart R. et al . Trocar-less instrumentation for laparoscopy magnetic positioning of intra-abdominal camera and retractor. Ann Surg. 2007; 245 379-384
- 16 Ryou M, Thompson C C. Magnetic retraction in natural-orifice transluminal endoscopic surgery (NOTES): addressing the problem of traction and countertraction. Endoscopy. 2010; 42 143-148
Scuola Superiore Sant’Anna
viale R. Piaggio 34