Subscribe to RSS
Intramuscular Near-Infrared Spectroscopy for Muscle Flap Monitoring in a Porcine Model
Background Current near-infrared spectroscopy (NIRS)-based systems for continuous flap monitoring are limited to flaps which carry a cutaneous paddle. As such, this useful and reliable technology has not previously been applicable to muscle-only free flaps where other modalities with substantial limitations continue to be utilized.
Methods We present the first NIRS probe which allows continuous monitoring of local tissue oxygen saturation (StO2) directly within the substance of muscle tissue. This probe is flexible, subcentimeter in scale, waterproof, biocompatible, and is fitted with resorbable barbs which facilitate temporary autostabilization followed by easy atraumatic removal. This novel device was compared with a ViOptix T.Ox monitor in a porcine rectus abdominus myocutaneous flap model of arterial and venous occlusions. During these experiments, the T.Ox device was affixed to the skin paddle, while the novel probe was within the muscle component of the same flap.
Results The intramuscular NIRS device and skin-mounted ViOptix T.Ox devices produced very similar StO2 tracings throughout the vascular clamping events, with obvious and parallel changes occurring upon vascular clamping and release. The normalized cross-correlation at zero lag describing correspondence between the novel intramuscular NIRS and T.Ox devices was >0.99.
Conclusion This novel intramuscular NIRS probe offers continuous monitoring of oxygen saturation within muscle flaps. This experiment demonstrates the potential suitability of this intramuscular NIRS probe for the task of muscle-only free flap monitoring, where NIRS has not previously been applicable. Testing in the clinical environment is necessary to assess durability and reliability.
* These authors contributed equally to this work.
Funding for this study was received from the Division of Plastic Surgery and the Department of Neurosurgery at Washington University, and from the Querry Simpson Institute of Bioelectronics at Northwestern University.
Received: 11 May 2021
Accepted: 31 May 2021
22 September 2021 (online)
© 2021. Thieme. All rights reserved.
Thieme Medical Publishers, Inc.
333 Seventh Avenue, 18th Floor, New York, NY 10001, USA
- 1 Chen K-T, Mardini S, Chuang DC-C. et al. Timing of presentation of the first signs of vascular compromise dictates the salvage outcome of free flap transfers. Plast Reconstr Surg 2007; 120 (01) 187-195
- 2 Chao AH, Meyerson J, Povoski SP, Kocak E. A review of devices used in the monitoring of microvascular free tissue transfers. Expert Rev Med Devices 2013; 10 (05) 649-660
- 3 Rozen WM, Enajat M, Whitaker IS, Lindkvist U, Audolfsson T, Acosta R. Postoperative monitoring of lower limb free flaps with the Cook-Swartz implantable Doppler probe: a clinical trial. Microsurgery 2010; 30 (05) 354-360
- 4 Guillemaud JP, Seikaly H, Cote D, Allen H, Harris JR. The implantable Cook-Swartz Doppler probe for postoperative monitoring in head and neck free flap reconstruction. Arch Otolaryngol Head Neck Surg 2008; 134 (07) 729-734
- 5 Zhang T, Dyalram-Silverberg D, Bui T, Caccamese Jr JF, Lubek JE. Analysis of an implantable venous anastomotic flow coupler: experience in head and neck free flap reconstruction. Int J Oral Maxillofac Surg 2012; 41 (06) 751-755
- 6 Rozen WM, Chubb D, Whitaker IS, Acosta R. The efficacy of postoperative monitoring: a single surgeon comparison of clinical monitoring and the implantable Doppler probe in 547 consecutive free flaps. Microsurgery 2010; 30 (02) 105-110
- 7 Berthelot M, Ashcroft J, Boshier P. et al. Use of near-infrared spectroscopy and implantable Doppler for postoperative monitoring of free tissue transfer for breast reconstruction: a systematic review and meta-analysis. Plast Reconstr Surg Glob Open 2019; 7 (10) e2437-e2438
- 8 Paydar KZ, Hansen SL, Chang DS, Hoffman WY, Leon P. Implantable venous Doppler monitoring in head and neck free flap reconstruction increases the salvage rate. Plast Reconstr Surg 2010; 125 (04) 1129-1134
- 9 Kagaya Y, Miyamoto S. A systematic review of near-infrared spectroscopy in flap monitoring: current basic and clinical evidence and prospects. J Plast Reconstr Aesthet Surg 2018; 71 (02) 246-257
- 10 Rosenberg JJ, Fornage BD, Chevray PM. Monitoring buried free flaps: limitations of the implantable Doppler and use of color duplex sonography as a confirmatory test. Plast Reconstr Surg 2006; 118 (01) 109-113 , discussion 114–115
- 11 Ferguson Jr REH, Yu P. Techniques of monitoring buried fasciocutaneous free flaps. Plast Reconstr Surg 2009; 123 (02) 525-532
- 12 Keller A. A new diagnostic algorithm for early prediction of vascular compromise in 208 microsurgical flaps using tissue oxygen saturation measurements. Ann Plast Surg 2009; 62 (05) 538-543
- 13 Koolen PGL, Vargas CR, Ho OA. et al. Does increased experience with tissue oximetry monitoring in microsurgical breast reconstruction lead to decreased flap loss? The learning effect. Plast Reconstr Surg 2016; 137 (04) 1093-1101
- 14 Bai W, Yang H, Ma Y. et al. Flexible transient optical waveguides and surface-wave biosensors constructed from monocrystalline silicon. Adv Mater 2018; 30 (32) e1801584
- 15 Bai W, Shin J, Fu R. et al. Bioresorbable photonic devices for the spectroscopic characterization of physiological status and neural activity. Nat Biomed Eng 2019; 3 (08) 644-654
- 16 Bodin F, Diana M, Koutsomanis A, Robert E, Marescaux J, Bruant-Rodier C. Porcine model for free-flap breast reconstruction training. J Plast Reconstr Aesthet Surg 2015; 68 (10) 1402-1409
- 17 Baker WB, Parthasarathy AB, Busch DR, Mesquita RC, Greenberg JH, Yodh AG. Modified Beer-Lambert law for blood flow. Biomed Opt Express 2014; 5 (11) 4053-4075
- 18 Sassaroli A, Fantini S. Comment on the modified Beer-Lambert law for scattering media. Phys Med Biol 2004; 49 (14) N255-N257
- 19 Kocsis L, Herman P, Eke A. The modified Beer-Lambert law revisited. Phys Med Biol 2006; 51 (05) N91-N98