Evaluation of the Implantable Doppler Probe for Free Flap Monitoring in Lower Limb ReconstructionFunding None of the authors have a financial interest in any of the products or devices mentioned in this article.
20 June 2017
28 September 2017
27 November 2017 (eFirst)
Background Timely reexploration and reanastomoses can salvage failing free flaps. The use of the implantable Doppler probe provides direct evidence of vascular impairment of the microvascular anastomoses and allows for postoperative NPWT. The aim of this retrospective study was to compare the Doppler probe to conventional monitoring techniques for free flap monitoring in lower limb reconstruction and to identify risk factors for perfusion disturbance and reexploration.
Methods All patients receiving free muscle flap reconstruction for lower limb soft tissue defects at our department from 2000 to 2013 were included, and all adverse events, timely detection of perfusion problems, and outcome of revision surgery were assessed by chart analysis.
Results For lower limb reconstruction, 110 free muscle transfers were performed of which 41 muscle flaps were conventionally monitored and 69 flaps were monitored using the implantable Doppler probe. In 18 cases, the free muscle flaps needed revision because of perfusion disturbances. The salvage rate was 80% with monitoring by the implantable Doppler probe compared with 62.5% using conventional monitoring methods resulting in success rates of 95.7 and 92.7%, respectively.
Conclusion The use of the implantable Cook–Swartz Doppler probe represents a safe monitoring method for lower limb reconstruction, which allows for the additional use of NPWT. Higher salvage and revision success rates can be attributed to an earlier detection of perfusion impairment. However, a larger patient cohort is necessary to verify superiority over conventional postoperative monitoring.
Keywordslower limb reconstruction - postoperative monitoring - implantable Doppler probe - free muscle flap - microsurgery
The study was approved by the ethics committee of the University of Freiburg Medical Center.
A part of this work was presented at the 38th Annual Conference of the German-Speaking Working Group for Peripheral Nerves and Vascular Microsurgery (DAM) held from December 3 to December 5, 2015, at Bochum, Germany.
- 1 Seidenberg B, Rosenak SS, Hurwitt ES, Som ML. Immediate reconstruction of the cervical esophagus by a revascularized isolated jejunal segment. Ann Surg 1959; 149 (02) 162-171
- 2 Chae MP, Rozen WM, Whitaker IS. , et al. Current evidence for postoperative monitoring of microvascular free flaps: a systematic review. Ann Plast Surg 2015; 74 (05) 621-632
- 3 Khouri RK, Shaw WW. Reconstruction of the lower extremity with microvascular free flaps: a 10-year experience with 304 consecutive cases. J Trauma 1989; 29 (08) 1086-1094
- 4 Benacquista T, Kasabian AK, Karp NS. The fate of lower extremities with failed free flaps. Plast Reconstr Surg 1996; 98 (05) 834-840 ; discussion 841–842
- 5 Culliford IV AT, Spector J, Blank A, Karp NS, Kasabian A, Levine JP. The fate of lower extremities with failed free flaps: a single institution's experience over 25 years. Ann Plast Surg 2007; 59 (01) 18-21 ; discussion 21–22
- 6 Kroll SS, Schusterman MA, Reece GP. , et al. Timing of pedicle thrombosis and flap loss after free-tissue transfer. Plast Reconstr Surg 1996; 98 (07) 1230-1233
- 7 Smit JM, Acosta R, Zeebregts CJ, Liss AG, Anniko M, Hartman EH. Early reintervention of compromised free flaps improves success rate. Microsurgery 2007; 27 (07) 612-616
- 8 Chen KT, Mardini S, Chuang DC. , et al. Timing of presentation of the first signs of vascular compromise dictates the salvage outcome of free flap transfers. Plast Reconstr Surg 2007; 120 (01) 187-195
- 9 Siemionow M, Arslan E. Ischemia/reperfusion injury: a review in relation to free tissue transfers. Microsurgery 2004; 24 (06) 468-475
- 10 Furnas H, Rosen JM. Monitoring in microvascular surgery. Ann Plast Surg 1991; 26 (03) 265-272
- 11 Whitaker IS, Gulati V, Ross GL, Menon A, Ong TK. Variations in the postoperative management of free tissue transfers to the head and neck in the United Kingdom. Br J Oral Maxillofac Surg 2007; 45 (01) 16-18
- 12 Jallali N, Ridha H, Butler PE. Postoperative monitoring of free flaps in UK plastic surgery units. Microsurgery 2005; 25 (06) 469-472
- 13 Neligan PC. Monitoring techniques for the detection of flow failure in the postoperative period. Microsurgery 1993; 14 (03) 162-164
- 14 Repez A, Oroszy D, Arnez ZM. Continuous postoperative monitoring of cutaneous free flaps using near infrared spectroscopy. J Plast Reconstr Aesthet Surg 2008; 61 (01) 71-77
- 15 Cai ZG, Zhang J, Zhang JG. , et al. Evaluation of near infrared spectroscopy in monitoring postoperative regional tissue oxygen saturation for fibular flaps. J Plast Reconstr Aesthet Surg 2008; 61 (03) 289-296
- 16 Mourouzis C, Anand R, Bowden JR, Brennan PA. Microdialysis: use in the assessment of a buried bone-only fibular free flap. Plast Reconstr Surg 2007; 120 (05) 1363-1366
- 17 Udesen A, Løntoft E, Kristensen SR. Monitoring of free TRAM flaps with microdialysis. J Reconstr Microsurg 2000; 16 (02) 101-106
- 18 Heller L, Levin LS, Klitzman B. Laser Doppler flowmeter monitoring of free-tissue transfers: blood flow in normal and complicated cases. Plast Reconstr Surg 2001; 107 (07) 1739-1745
- 19 Hölzle F, Loeffelbein DJ, Nolte D, Wolff KD. Free flap monitoring using simultaneous non-invasive laser Doppler flowmetry and tissue spectrophotometry. J Craniomaxillofac Surg 2006; 34 (01) 25-33
- 20 Liss AG, Liss P. Use of a modified oxygen microelectrode and laser-Doppler flowmetry to monitor changes in oxygen tension and microcirculation in a flap. Plast Reconstr Surg 2000; 105 (06) 2072-2078
- 21 Swartz WM, Jones NF, Cherup L, Klein A. Direct monitoring of microvascular anastomoses with the 20-MHz ultrasonic Doppler probe: an experimental and clinical study. Plast Reconstr Surg 1988; 81 (02) 149-161
- 22 Swartz WM, Izquierdo R, Miller MJ. Implantable venous Doppler microvascular monitoring: laboratory investigation and clinical results. Plast Reconstr Surg 1994; 93 (01) 152-163
- 23 Bannasch H, Iblher N, Penna V. , et al. A critical evaluation of the concomitant use of the implantable Doppler probe and the vacuum assisted closure system in free tissue transfer. Microsurgery 2008; 28 (06) 412-416
- 24 Eisenhardt SU, Momeni A, Iblher N. , et al. The use of the vacuum-assisted closure in microsurgical reconstruction revisited: application in the reconstruction of the posttraumatic lower extremity. J Reconstr Microsurg 2010; 26 (09) 615-622
- 25 Eisenhardt SU, Schmidt Y, Thiele JR. , et al. Negative pressure wound therapy reduces the ischaemia/reperfusion-associated inflammatory response in free muscle flaps. J Plast Reconstr Aesthet Surg 2012; 65 (05) 640-649
- 26 Iblher N, Eisenhardt SU, Penna V, Stark GB, Bannasch H. A new evaluation tool for monitoring devices and its application to evaluate the implantable Doppler probe. J Reconstr Microsurg 2010; 26 (04) 265-270
- 27 Rozen WM, Enajat M, Whitaker IS, Lindkvist U, Audolfsson T, Acosta R. Postoperative monitoring of lower limb free flaps with the Cook-Swartz implantable Doppler probe: a clinical trial. Microsurgery 2010; 30 (05) 354-360
- 28 Smit JM, Werker PM, Liss AG. , et al. Introduction of the implantable Doppler system did not lead to an increased salvage rate of compromised flaps: a multivariate analysis. Plast Reconstr Surg 2010; 125 (06) 1710-1717
- 29 Kind GM, Buntic RF, Buncke GM, Cooper TM, Siko PP, Buncke Jr HJ. The effect of an implantable Doppler probe on the salvage of microvascular tissue transplants. Plast Reconstr Surg 1998; 101 (05) 1268-1273 ; discussion 1274–1275
- 30 Rozen WM, Chubb D, Whitaker IS, Acosta R. The efficacy of postoperative monitoring: a single surgeon comparison of clinical monitoring and the implantable Doppler probe in 547 consecutive free flaps. Microsurgery 2010; 30 (02) 105-110
- 31 Frost MW, Niumsawatt V, Rozen WM, Eschen GE, Damsgaard TE, Kiil BJ. Direct comparison of postoperative monitoring of free flaps with microdialysis, implantable cook-swartz Doppler probe, and clinical monitoring in 20 consecutive patients. Microsurgery 2015; 35 (04) 262-271
- 32 Schmulder A, Gur E, Zaretski A. Eight-year experience of the Cook-Swartz Doppler in free-flap operations: microsurgical and reexploration results with regard to a wide spectrum of surgeries. Microsurgery 2011; 31 (01) 1-6
- 33 Rosenberg JJ, Fornage BD, Chevray PM. Monitoring buried free flaps: limitations of the implantable Doppler and use of color duplex sonography as a confirmatory test. Plast Reconstr Surg 2006; 118 (01) 109-113 ; discussion 114–115
- 34 de la Torre J, Hedden W, Grant III JH, Gardner PM, Fix RJ, Vásconez LO. Retrospective review of the internal Doppler probe for intra- and postoperative microvascular surveillance. J Reconstr Microsurg 2003; 19 (05) 287-290
- 35 Pryor SG, Moore EJ, Kasperbauer JL. Implantable Doppler flow system: experience with 24 microvascular free-flap operations. Otolaryngol Head Neck Surg 2006; 135 (05) 714-718
- 36 Smit JM, Zeebregts CJ, Acosta R, Werker PM. Advancements in free flap monitoring in the last decade: a critical review. Plast Reconstr Surg 2010; 125 (01) 177-185
- 37 Dornseifer U, Kleeberger C, Kargl L. , et al. Perfusion controlled mobilization after lower extremity free flaps-pushing the limits of time and intensity. J Reconstr Microsurg 2017; 33 (03) 179-185
- 38 Neubert N, Vogt PM, May M. , et al. Does an early and aggressive combined wrapping and dangling procedure affect the clinical outcome of lower extremity free flaps?-A randomized controlled prospective study using microdialysis monitoring. J Reconstr Microsurg 2016; 32 (04) 262-270
- 39 Kolbenschlag J, Bredenbroeker P, Lehnhardt M. , et al. Advanced microcirculatory parameters of lower extremity free flaps during dangling and their influencing factors. J Reconstr Microsurg 2015; 31 (07) 500-507