Automated Smartphone Threshold Audiometry: Validity and Time Efficiency
26 June 2020 (online)
Smartphone-based threshold audiometry with automated testing has the potential to provide affordable access to audiometry in underserved contexts.
To validate the threshold version (hearTest) of the validated hearScreen™ smartphone-based application using inexpensive smartphones (Android operating system) and calibrated supra-aural headphones.
A repeated measures within-participant study design was employed to compare air-conduction thresholds (0.5–8 kHz) obtained through automated smartphone audiometry to thresholds obtained through conventional audiometry.
A total of 95 participants were included in the study. Of these, 30 were adults, who had known bilateral hearing losses of varying degrees (mean age = 59 yr, standard deviation [SD] = 21.8; 56.7% female), and 65 were adolescents (mean age = 16.5 yr, SD = 1.2; 70.8% female), of which 61 had normal hearing and the remaining 4 had mild hearing losses.
Threshold comparisons were made between the two test procedures. The Wilcoxon signed-ranked test was used for comparison of threshold correspondence between manual and smartphone thresholds and the paired samples t test was used to compare test time.
Within the adult sample, 94.4% of thresholds obtained through smartphone and conventional audiometry corresponded within 10 dB or less. There was no significant difference between smartphone (6.75-min average, SD = 1.5) and conventional audiometry test duration (6.65-min average, SD = 2.5). Within the adolescent sample, 84.7% of thresholds obtained at 0.5, 2, and 4 kHz with hearTest and conventional audiometry corresponded within ≤5 dB. At 1 kHz, 79.3% of the thresholds differed by ≤10 dB. There was a significant difference (p < 0.01) between smartphone (7.09 min, SD = 1.2) and conventional audiometry test duration (3.23 min, SD = 0.6).
The hearTest application with calibrated supra-aural headphones provides a cost-effective option to determine valid air-conduction hearing thresholds.
- Abu-Ghanem S, Handzel O, Ness L, Ben-Artzi-Blima M, Fait-Ghelbendorf K, Himmelfarb M. 2015; Smartphone-based audiometric test for screening hearing loss in the elderly. Eur Arch Otorhinolaryngol 273 (02) 333-339
- American Speech-Language-Hearing Association (ASHA) 2011 Type, degree and configuration of hearing loss. http://www.asha.org/uploadedFiles/AIS-Hearing-Loss-Types-Degree-Configuration.pdf . Accessed June 8, 2015.
- Burdette SD, Herchline TE, Oehler R. 2008; Surfing the web: practicing medicine in a technological age: using smartphones in clinical practice. Clin Infect Dis 47 (01) 117-122
- Chin R, Lee B. 2008. Principles and Practice of Clinical Trial Medicine. London, UK: Elsevier Inc;
- Clark JL, Swanepoel W. 2014; Technology for hearing loss—as We Know it, and as We Dream it. Disabil Rehabil Assist Technol 9 (05) 408-413
- Duggal P, Sarkar M. 2007; Audiologic monitoring of multi-drug resistant tuberculosis patients on aminoglycoside treatment with long term follow-up. BMC Ear Nose Throat Disord 7 (05) 5
- Eikelboom RH, Swanepoel DW, Motakef S, Upson GS. 2013; Clinical validation of the AMTAS automated audiometer. Int J Audiol 52 (05) 342-349
- Fagan JJ, Jacobs M. 2009; Survey of ENT services in Africa: need for a comprehensive intervention. Glob Health Action 2: 1-7
- Foulad A, Bui P, Djalilian H. 2013; Automated audiometry using Apple iOS-based application technology. Otolaryngol Head Neck Surg 149 (05) 700-6
- Goulios H, Patuzzi RB. 2008; Audiology education and practice from an international perspective. Int J Audiol 47 (10) 647-664
- Harris T, Peer S, Fagan J. 2012; Audiological monitorig for ototoxic tuberculosis, human immunodeficiency virus and cancer therapies in a developing world setting. J Laryngol Oto 126: 548-551
- Ho A, Hildreth A, Lindsey L. 2009; Computer-assisted audiometry versus manual audiometry. Otol Neurotol 30: 876-883
- International Standards Organization (ISO). (1983) Acoustics—Pure Tone Air Conduction Threshold Audiometry for Hearing Conservation Purposes (ISO 6189:1983)
- International Standards Organization (ISO). (1994) Acoustics—Reference Zero for the Calibration of Audiometric Equipment—Part 2: Reference Equivalent Threshold Sound Pressure Levels for Pure Tones and Insert Earphones (ISO 389-2:1994)
- International Standards Organization (ISO). (1998) Acoustics—Reference Zero for the Calibration of Audiometric Equipment—Part 1: Reference Equivalent Thresholds Sound Pressure Levels for Pure Tones and Circumaural Earphones (ISO 389-1:1998)
- International Standards Organization (ISO). (2009) Acoustics—Reference Zero for the Calibration of Audiometric Equipment—Part 9: Preferred Test Conditions for the Determination of Reference Hearing Threshold Levels (ISO 389-9:2009)
- International Standards Organization (ISO). (2010) Acoustics—Audiometric Test Methods—Part 1: Pure Tone Air and Bone Conduction Audiometry (ISO 8253-1:2010)
- Kelly T, Minges M. 2012. Exclusive Summary. Washington, DC: World Bank;
- Khoza-Shangase K, Kassner L. 2013; Automated screening audiometry in the digital age: exploring uHear™ and its use in a resource-stricken developing country. Int J Technol Assess Health Care 29 (01) 42-47
- Kochi E. 2012 How the future of mobile lies in the developing world. http://techcrunch.com/2012/05/27/mobile-developing-world/ . Accessed February 27, 2016
- Leedy P, Ormrod J. 2014. Practical Research: Planning and Design. 10th ed. Essex, UK: Pearson Education Limited;
- Mahomed F, Swanepoel D, Eikelboom RH, Soer M. 2013; Validity of automated threshold audiometry: a systematic review and meta-analysis. Ear Hear 34 (06) 745-752
- Mahomed-Asmail F, Swanepoel D, Eikelboom RH. 2016; Diagnostic hearing assessment in schools: validity and time-efficiency of automated audiometry. J Am Acad Audiol 27 (01) 42-48
- Mahomed-Asmail F, Swanepoel D, Eikelboom R, Myburgh H, Hall J. 2015; Clinical validity of hearScreen smartphone: hearing screening for school children. Ear Hear 37 (01) e11-e17
- Margolis RH, Morgan DE. 2008; Automated pure-tone audiometry: an analysis of capacity, need, and benefit. Am J Audiol 17 (02) 109-113
- McDaniel M, Chinn G, McCall K, Stewart A. 2013 Guidelines for audiometric baseline revision. http://c.ymcdn.com/sites/www.hearingconservation.org/resource/resmgr/Position_Statements/Guidelines_for_Audiometric_B.pdf . Accessed February 27, 2016
- Mosa AS, Yoo I, Sheets L. 2012; A systematic review of healthcare applications for smartphones. BMC Med Inform Decis Mak 12: 67-98
- Occupational Saftey and Health Administration (OSHA) 1983; Occupational noise exposure; hearing conservation amendment. Fed Regist 48: 9738-9785
- Peer S, Fagan JJ. 2015; Hearing loss in the developing world: evaluating the iPhone mobile device as a screening tool. S Afr Med J 105 (01) 35-39
- Swanepoel D, Biagio L. 2011; Validity of diagnostic computer-based air and forehead bone conduction audiometry. J Occup Environ Hyg 8 (04) 210-214
- Swanepoel D, Clark JL, Koekemoer D, Hall 3rd JW, Krumm M, Ferrari DV, McPherson B, Olusanya BO, Mars M, Russo I, Barajas JJ. 2010; Telehealth in audiology: the need and potential to reach underserved communities. Int J Audiol 49 (03) 195-202
- Swanepoel D, Maclennan-Smith F, Hall J. 2013; Diagnostic pure-tone audiometry in schools: Mobile testing without a sound-treated evironment. J Am Acad Audiol 24: 992-1000
- Swanepoel D, Mngemane S, Molemong S, Mkwanazi H, Tutshini S. 2010; Hearing assessment—reliability, accuracy, and efficiency of automated audiometry. Telemed J E Health 16 (05) 557-563
- Swanepoel D, Myburgh H, Howe D, Mahomed F, Eikelboom R. 2014; Smartphone hearing screening with integrated quality control and data management. Int J Audiol 53 (12) 841-9
- Swanepoel D, Olusanya BO, Mars M. 2010; Hearing health-care delivery in sub-Saharan Africa—a role for tele-audiology. J Telemed Telecare 16 (02) 53-56
- Szudek J, Ostevik A, Dziegielewski P, Robinson-Anagor J, Gomaa N, Hodgetts B, Ho A. 2012; Can Uhear me now? Validation of an iPod-based hearing loss screening test. J Otolaryngol Head Neck Surg 41: 78-84
- Thompson G, Sladen D, Borst BJ, Still O. 2015; Accuracy of a tablet audiometer for measuring behavioural hearing thresholds in a clinical population. Otolaryngology Head Neck Surg 153 (05) 838-842
- Van der Aerschot M, Swanepoel D, Mahomed-Asmail F, Myburgh HC, Eikelboom RH. 2016 Affordable headphones for accessible screening audiometry: an evaluation of the Sennheiser HD202 II supra-aural headphone. Int J Audiol. [in press]
- Windmill IM, Freeman BA. 2013; Demand for audiology services: 30-yr projections and impact on academic programs. J Am Acad Audiol 24 (05) 407-416
- World Health Organisation (WHO) 2011. mHealth: New Horizons for Health through Mobile Technologies: Second Global Survey on eHealth. Geneva, Switzerland: World Health Organisation;
- World Health Organisation (WHO) 2013. Multi-Country Assessment of National Capacity to Provide Hearing Care. Geneva, Switzerland: World Health Organisation;
- World Health Organisation (WHO) 2014 Fact sheet 300: Deafness and hearing impairment. http://0-www.who.int.innopac.up.ac.za/mediacentre/factsheets/fs300/en/ . Accessed June 8, 2015
- Yeung J, Javidnia H, Heley S, Beauregard Y, Champagne S, Bromwich M. 2013; The new age of play audiometry: prospective validation testing of an iPad-based play audiometer. J Otolaryngol Head Neck Surg 42: 21
- Yeung JC, Heley S, Beauregard Y, Champagne S, Bromwich MA. 2015; Self-administered hearing loss screening using an interactive, tablet play audiometer with ear bud headphones. Int J Pediatr Otorhinolaryngol 79 (08) 1248-1252
- Yousuf Hussein S, Swanepoel W, Biagio de Jager L, Myburgh HC, Eikelboom RH, Hugo J. 2015; Smartphone hearing screening in mHealth assisted community-based primary care. J Telemed Telecare [early online edition] 1-8