Detecting Microsurgical Complications with ViOptix Tissue Oximetry in a Pediatric Myocutaneous Free Flap: Case Presentation and Literature Review
16 November 2017
23 December 2017
12 February 2018 (online)
Background Microvascular compromise from arterial or venous occlusion is a common cause of free flap failure. The salvage rate following a microvascular compromise is dependent on detecting the problem early and intervening quickly.
Methods The ViOptix tissue oximeter measures tissue oxygen saturation using the near-infrared spectroscopy technology. The ViOptix device has an alarm capability to warn of potential compromise to tissue perfusion. The tissue oximetry readings are visible on the bedside monitor and are relayed to a webpage link, which is accessible on a personal computer or mobile device, allowing real-time monitoring. This article presents a case where real-time monitoring allowed almost immediate detection of inadvertent pedicle compromise allowing flap salvage by repositioning without surgical intervention.
Results In the case presented, the patient's nurse inadvertently positioned a pillow under the location of the vascular pedicle likely causing microvascular compression. The ViOptix reading dropped and for this reason the nurse contacted the Plastic Surgery team. The drop was confirmed remotely and the flap was urgently evaluated in person. Once the pillow was removed, the ViOptix readings normalized and Doppler signals strengthened in the flap.
Discussion While tissue oximetry monitoring does not by itself ensure flap survival, it provides critical information than conventional flap monitoring would allow giving the microsurgeon the opportunity to make a quicker decision. ViOptix tissue oximeters are able to detect vascular compromise even before conventional clinical symptoms are present. Alas in several cases by the time clinical symptoms develop the flap may be beyond salvage.
- 1 Wei FC, Mardini S. Flaps and Reconstructive Surgery. 2nd ed. Chinca: Elsevier Health Sciences; 2016
- 2 Khouri RK, Cooley BC, Kunselman AR. , et al. A prospective study of microvascular free-flap surgery and outcome. Plast Reconstr Surg 1998; 102 (03) 711-721
- 3 Koolen PG, Vargas CR, Ho OA. , et al. Does increased experience with tissue oximetry monitoring in microsurgical breast reconstruction lead to decreased flap loss? The learning effect. Plast Reconstr Surg 2016; 137 (04) 1093-1101
- 4 Keller A. Noninvasive tissue oximetry for flap monitoring: an initial study. J Reconstr Microsurg 2007; 23 (04) 189-197
- 5 Colwell AS, Craft RO. Near-infrared spectroscopy in autologous breast reconstruction. Clin Plast Surg 2011; 38 (02) 301-307
- 6 Lin SJ, Nguyen MD, Chen C. , et al. Tissue oximetry monitoring in microsurgical breast reconstruction decreases flap loss and improves rate of flap salvage. Plast Reconstr Surg 2011; 127 (03) 1080-1085
- 7 Bank J, Beederman M, Shore AM, Song DH. Mechanical leeching with venocutaneous fistula and monitoring with near-infrared spectroscopy. Plast Reconstr Surg Glob Open 2013; 1 (07) e56
- 8 Keller A. A new diagnostic algorithm for early prediction of vascular compromise in 208 microsurgical flaps using tissue oxygen saturation measurements. Ann Plast Surg 2009; 62 (05) 538-543
- 9 Steele MH. Three-year experience using near infrared spectroscopy tissue oximetry monitoring of free tissue transfers. Ann Plast Surg 2011; 66 (05) 540-545
- 10 Lin CM, Huang CC, Hsu H, Chiu CH, Chien SH. Advancements in free flap monitoring in the last decade: a critical review. Plast Reconstr Surg 2010; 126 (02) 679 , author reply 679–680
- 11 Scheufler O, Exner K, Andresen R. Investigation of TRAM flap oxygenation and perfusion by near-infrared reflection spectroscopy and color-coded duplex sonography. Plast Reconstr Surg 2004; 113 (01) 141-152 , discussion 153–155
- 12 Colwell AS, Wright L, Karanas Y. Near-infrared spectroscopy measures tissue oxygenation in free flaps for breast reconstruction. Plast Reconstr Surg 2008; 121 (05) 344e-345e
- 13 Repez A, Oroszy D, Arnez ZM. Continuous postoperative monitoring of cutaneous free flaps using near infrared spectroscopy. J Plast Reconstr Aesthet Surg 2008; 61 (01) 71-77
- 14 Bellamy JL, Mundinger GS, Flores JM. , et al. Do adjunctive flap-monitoring technologies impact clinical decision making? An analysis of microsurgeon preferences and behavior by body region. Plast Reconstr Surg 2015; 135 (03) 883-892
- 15 Smit JM, Zeebregts CJ, Acosta R, Werker PM. Advancements in free flap monitoring in the last decade: a critical review. Plast Reconstr Surg 2010; 125 (01) 177-185
- 16 Pelletier A, Tseng C, Agarwal S, Park J, Song D. Cost analysis of near-infrared spectroscopy tissue oximetry for monitoring autologous free tissue breast reconstruction. J Reconstr Microsurg 2011; 27 (08) 487-494