Comparison of Locking and Frag-Loc Screws for Fixation of Die-Punch FragmentsFunding This work is supported by Department of Anatomy, School of Medicine, Showa University and Department of Mechanical and Production Engineering, Niigata University.
07 August 2017
07 November 2017
14 December 2017 (eFirst)
Background The Frag-Loc (FL) compression screw system was designed to stabilize dorsally displaced intra-articular dorsoulnar (die-punch) fragments in distal radius fractures.
Purpose Comparison of the biomechanical properties of fixation of the die-punch fragment (stiffness, ultimate strength, and displacement ratio of the fragment), using the FL and traditional locking screw (LS), and using simulated distal radial fractures in cadaveric specimens under axial compressive loading. Both screws were used with a volar locking plate (VLP).
Materials and Methods Eight matched pairs of formalin-fixed cadaveric specimens of the radius were used to simulate distal radius fractures with die-punch fragments. The die-punch fragment was fixed using VLP with either FL group or LS group. Biomechanical properties for the two fixation systems were evaluated under axial compression loading, applied at a constant rate of 0.5 mm/min until failure. Load data were recorded and the ultimate strength and change in the gap between the die-punch and proximal fragments measured, with the displacement ratio calculated by dividing the value of the gap before loading by the gap after loading. Failure was defined as 10 mm or more of fragment displacement, or screw failure.
Results There were no differences in ultimate strength (p = 0.47) or stiffness (p = 0.061) between the two fragment fixation systems. However, the displacement ratio was lower for the FL than for the LS system (p = 0.049).
Conclusion Compared with LS, the FL system lowers the displacement of die-punch fragments under compressive loading.
Clinical Relevance The FL system is effective for the treatment of distal radius fractures with die-punch fragments.
Keywordsdie-punch (dorsoulnar) fragment - displacement ratio - distal radius fracture - Frag-Loc screw - locking screw
All procedures performed in studies involving human participants were in accordance with the ethical standards and with the 1964 Helsinki declaration and its later amendments or comparable ethical standards.
- 1 Chung KC, Watt AJ, Kotsis SV, Margaliot Z, Haase SC, Kim HM. Treatment of unstable distal radial fractures with the volar locking plating system. J Bone Joint Surg Am 2006; 88 (12) 2687-2694
- 2 Jupiter JB, Marent-Huber M. ; LCP Study Group. Operative management of distal radial fractures with 2.4-millimeter locking plates. A multicenter prospective case series. J Bone Joint Surg Am 2009; 91 (01) 55-65
- 3 Chang HC, Poh SY, Seah SC, Chua DTC, Cha BK, Low CO. Fragment-specific fracture fixation and double-column plating of unstable distal radial fractures using AO mini-fragment implants and Kirschner wires. Injury 2007; 38 (11) 1259-1267
- 4 Kamano M, Honda Y, Kazuki K, Yasuda M. Palmar plating for dorsally displaced fractures of the distal radius. Clin Orthop Relat Res 2002; (397) 403-408
- 5 Arora R, Lutz M, Fritz D, Zimmermann R, Oberladstätter J, Gabl M. Palmar locking plate for treatment of unstable dorsal dislocated distal radius fractures. Arch Orthop Trauma Surg 2005; 125 (06) 399-404
- 6 Osada D, Kamei S, Masuzaki K, Takai M, Kameda M, Tamai K. Prospective study of distal radius fractures treated with a volar locking plate system. J Hand Surg Am 2008; 33 (05) 691-700
- 7 Lee JI, Cho JH, Lee SJ. The effects of the Frag-Loc(®) compression screw on distal radius fracture with a displaced dorsoulnar fragment. Arch Orthop Trauma Surg 2015; 135 (09) 1315-1321
- 8 Ikeda K, Osamura N, Tada K. Fixation of an ulnodorsal fragment when treating an intra-articular fracture in the distal radius. Hand Surg 2014; 19 (01) 139-144
- 9 Ruch DS, Papadonikolakis A. Volar versus dorsal plating in the management of intra-articular distal radius fractures. J Hand Surg Am 2006; 31 (01) 9-16
- 10 Kim JK, Cho SW. The effects of a displaced dorsal rim fracture on outcomes after volar plate fixation of a distal radius fracture. Injury 2012; 43 (02) 143-146
- 11 Rausch S, Schlonski O, Klos K. , et al. Volar versus dorsal latest-generation variable-angle locking plates for the fixation of AO type 23C 2.1 distal radius fractures: a biomechanical study in cadavers. Injury 2013; 44 (04) 523-526
- 12 Carter PR, Frederick HA, Laseter GF. Open reduction and internal fixation of unstable distal radius fractures with a low-profile plate: a multicenter study of 73 fractures. J Hand Surg Am 1998; 23 (02) 300-307
- 13 Kim JK, Yun YH, Kim DJ. The effect of displaced dorsal rim fragment in a distal radius fracture. J Wrist Surg 2016; 5 (01) 31-35
- 14 Osada D, Fujita S, Tamai K, Iwamoto A, Tomizawa K, Saotome K. Biomechanics in uniaxial compression of three distal radius volar plates. J Hand Surg Am 2004; 29 (03) 446-451
- 15 Johnson NA, Cutler L, Dias JJ, Ullah AS, Wildin CJ, Bhowal B. Complications after volar locking plate fixation of distal radius fractures. Injury 2014; 45 (03) 528-533
- 16 Moss DP, Means Jr KR, Parks BG, Forthman CL. A biomechanical comparison of volar locked plating of intra-articular distal radius fractures: use of 4 versus 7 screws for distal fixation. J Hand Surg Am 2011; 36 (12) 1907-1911
- 17 Liporace FA, Kubiak EN, Jeong GK, Iesaka K, Egol KA, Koval KJ. A biomechanical comparison of two volar locked plates in a dorsally unstable distal radius fracture model. J Trauma 2006; 61 (03) 668-672